Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||25 July 2013|
|PDF File Size:||17.1 Mb|
|ePub File Size:||1.70 Mb|
|Price:||Free* [*Free Regsitration Required]|
Agilent / HP 4191A RF Impedance Analyzer
Message 6 of 6. Message 1 of 6. Message 5 of 6. Accepted by topic author Matt C.
IIRC, the sweep parameters were something like this: Thanks for your help guys. Message 4 of 6. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: Good luck with it I think the would probably take the old commands, but the ALSO has new commands.
In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
A RF IMPEDANCE ANL [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen. The forums emoticons took over for a bit there.
Message 2 of 6. Should I be looking at doing it with Hpp4191a or something else? If not, what would creating a driver for it entail?
HP A for Sale|Impedance Analyzers|Impedance Measurement Products|Test Equipment Center
I have not worked with that instrument, but maanual Agilent web site has the operating and service manual here. When I gave it a try, though, it worked fine. I suspect that this difference might be causing my problems. Guess you never know until you try! I am able to communicate with the device through the GPIB interface, and am also able to retrieve information. Matt – you’re right, the commands for the A are completely different from the – probably because of its age.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
4191A RF IMPEDANCE ANL
Would you mind posting the code you used on the A? You really need the manual. Message 3 of 6. In the code above, MHz is an arbitrary frequency.
It’s 84MB, so I hope you have a fast link. I went through the manual before, but when Mannual saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to ph4191a different. Sorry, that last line of code there is supposed to read: Most Active Software Boards: However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at.
Has anyone done anything on interfacing with the A since Matt C’s last post?