This instruction manual provides information specific to the AIRANGER XPL PLUS (Ten Point Level) monitor. When references are made to other Milltronics. AiRanger XPL Plus. * Program range is defined as the empty distance to the face of the transducer (P) plus any range extension (P). **EMC performance. AiRanger XPL Plus. Ten-Point Ultrasonic Level Measurement System. Part 1. General. Scope. A. This section describes the requirements for an ultrasonic.
|Published (Last):||7 July 2014|
|PDF File Size:||6.67 Mb|
|ePub File Size:||9.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
For superior air temperature compensation, a TIB-9 card may be used with individual Milltronics TS-3 temperature sensors for each vessel.
A simple calibration feature compensates for homogeneous consistent composition atmospheres other than air. RS or RS interface. Numerous failsafe options for process control equipment activation. Based on level, space, distance, or volume, Application Examples for additional installation requirements. Follow proper grounding procedures. Loosen the 6 enclosure lid captivated screws and swing the lid open.
Remove the 4 board B mounting screws outer xlp and remove the circuit board assembly.
Milltronics Sam, Airanger XPL Plus, Milltronics BIC II | eBay
To avoid differential ground potentials, do not connect cable shields to ground earth elsewhere. Insulate tape cable shields at all shield junctions to prevent ground loops. Not all components or their maximum quantity may be required. Transducer cables must be run in a grounded metal conduit separate from other wiring except TS-3 temperature sensor wiring, if applicable. Hazardous voltage present on transducer terminals during operation.
Ensure that any connected alarm or control equipment is disabled until satisfactory system operation and performance is verified. To enter the program mode Ensure the enclosure lid is secured by the 6 captivated screws.
Place the infrared programmer in the enclosure lid recess no wiring or fastening required. When the program mode is entered after RUN mode operation, all operating data is retained in memory.
In many cases, when a Parameter Value is altered, associated Parameter Values are automatically altered accordingly. When a Parameter Number is accessed, if the preset Parameter Value displayed is acceptable, no entry is required. To return an operator adjusted Parameter Value to the preset value, with the appropriate Point Number and Parameter Number displayed When a Transducer P value is entered for Point 1 only, the Reading and other data is displayed continuously.
If Volume Conversion is not required, proceed to Reading Parameters. Enter the vessel volume between Empty P and Span P This feature is preset to 0. To reference a material level to sea level, enter the distance in Units Pbetween Empty P and sea level.
Siemens SITRANS LU10 Ten-Point Ultrasonic Level Transmitter (Milltronics AiRanger XPL Plus)
Enter a negative value if Empty plsu below sea level. To operate process control equipment under these conditions, alter the following parameters as required. Relays may be programmed as Standard Alarms or for Custom Relay operation. P 4 mA TRIM Use this feature in combination with 20 mA Trim if the device connected to the mA output displayed is out of calibration, yet device recalibration is impractical. Adjust this value as required to make the device connected indicate 4.
Use these features to record and save a total of up to 10 Echo profiles, initiated manually Por automatically P et al. To save a manual record: The parameter value field displays the new record ordinates. Before using this feature, verify the: This value is automatically adjusted when an Offset Calibration is performed.
Alternatively, if the amount of Offset Correction required is known and performing an Offset Calibration is not desiredenter the amount to be plys to the Reading before display. To alter this allocation, enter the TS-3 whose temperature measurement will be used for the distance calculation of the Point Number displayed. When more than one TS-3 temperature sensor is allocated to a Point Number, the temperature measurements from each sensor are averaged for that vessel.
Enter a value slightly greater than the maximum vessel filling rate. This value is automatically altered when Measurement Response P is altered. P value P preset airsnger 1, max verification In the RUN mode, if any of the conditions occur, Priority Point s are scanned each time a non- priority point is scanned.
Enter the amount of delay in seconds. Refer to the module documentation for a list and description of the specific parameter requirements. Extend Near Blanking to overcome measurement difficulties which cannot be corrected by transducer location, mounting, or aiming. The causes of measurement difficulties which may be corrected include: Select “short and long” to have short and long acoustic shots fired each time a vessel is scanned.
Select “short” to have only short shots fired if the Echo Confidence P produced by a short shot exceeds the short Confidence Threshold P The Profile Pointer may be moved to a number of points on the Echo Profile, to gain specific information dependent upon the Profile Pointer Parameter used.
To move the Profile Pointer to a specific point, enter the desired value. This value is automatically altered by Material P If the wrong echo is processed, select an alternate algorithm, while observing the resultant echo processing displays.
Enter the amount in ms of long shot Echo Profile smoothing required to eliminate jagged peaks in the Echo Profile. To select another transducer. Select desired transducer point. This feature is preset to Before activating this feature, connect terminal block contacts 42 to 45 and 43 to On successful test completion, “PASS” is displayed. If “FAIL” is displayed, zpl the test.
The first test performed sets up the auto polarity function. Access the following parameters to take an ultrasonic measurement and: Enter the point number to copy from. To copy a parameter set. Following a Global Master Reset, complete reprogramming is required. The transducer fires an acoustic “shot” Unless altered, the sound velocity used for the distance calculation is If the transducer airabger are exposed to direct sunlight, TS-3 temperature sensor s should be used.
If the vessel to be monitored does not match any of the 8 preset Tank Shape calculations, a Universal Volume calculation may be used. Enter “2” for solid. No entry required, preset for metres. xpo
Milltronics airanger XPL plus Instruction Manual
No entry xpll preset for liquid. P Units Enter “4” As well as monitoring the amount of material in each bunker section or tank a mA output representing the average measurement may be used to calculate total material or remaining capacity. Operation is similar to Space Measurement operation except that measurements are referenced to the transducer face. Data messages are continuously transmitted at regular intervals no poll is required at baud. Wipe out the area of the enclosure lid docking bay recess with airsnger clean dry cloth if necessary before installing the programmer.
No response to programmer. Obstructed infrared interface, or Clean enclosure “docking bay” defective programmer. Transducer sound beam is free from obstruction. If the material monitored is a liquid, check for splashing in the vessel. Increase Measurement Response as required to stabilize the Reading, or install a stilling well contact Milltronics or your local distributor.
Access the Noise P parameter. If the peak value jumps wildly, verify the transducer cables are run in grounded metal conduit; Refer to associated product documentation.
Page of Go. Table of Contents Add to my manuals Add. Installation Application Examples for additional installation requirements.
Page 46 P 4 mA TRIM Use this feature in combination xxpl 20 mA Trim if the device connected to the mA output displayed is out of calibration, yet device recalibration is impractical. Page 52 To save a manual record: Page 68 Extend Near Blanking to overcome measurement difficulties which cannot be corrected by transducer location, mounting, or aiming. Page 75 enters the change in value. Distance Calculation Unless altered, the sound velocity used for the distance calculation is Wrong Reading If the material monitored is a liquid, check for splashing in the vessel.
Page Cable Transducer: Comments to this Manuals Your Name. Enter text from picture: By pressing ‘print’ button you will print only current page.